ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
|Published (Last):||28 September 2010|
|PDF File Size:||11.66 Mb|
|ePub File Size:||5.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
To toggle between two options : Fill the floorpan of the work space with a sufficient quantity of water.
Page Circuit breaker G 16 0. My Bench Order History Sign out.
Thermo Scientific Heracell i Manuals
Position the new seal at a corner  and press the seal retaining rail  into the slot. Information on these tests is available from Thermo Scientific upon request.
To resume the auto-start routine: Chapter 10 Maintenance Note Isothermal container: Note Configuration with split inserts: Notice The contra-con decontamination routine must be performed every 3 months. The auto-start routine is restarted automatically if the routine is canceled due to one of the following events: Support hook for shelf Page Glass door seal, measurement opening Tempered silicone Sensor block Stainless steel 1.
Thermo HERAcell-CO2-Incubator 150i 240i manual en
Page If both bottles are empty: Page 31 independent of device type Water pump Water pump Figure. Remove all samples an brought in tools from the device, manuap clean and disinfect them. Exit the error dialog: Kanual have one of these CO2 inbators that has condensation building up around seal on the inside of the glass door. Page 52 i may be equipped with up to four separate bottle turning devices. Measuring cell with blower wheel and sensors 5.
To exit the error display: Installing The Shelf System Tools are not required for the installation of the shelf system. The center strut is stabilized by the spring pressure. Each unit consists CELL of a roller insert  with a drive roller  and four idle rollers  that are also rotated due to the transmission 240l of the inserted cylinders .
Thermo Scientific Heracell 240i Operating Instructions Manual
Table Of Contents Replacing the door seal Page Failure to set the gas-tight screen function in accordance with the actually installed door version may impair the incubation performance.
Page 49 The outer door of the device is also heated. To accept and save the setting: The cross ueracell of the openings are smaller when samples are accessed.
Access port with plug Actual value CO 8. Latch, glass door This manual also for: You can adjust your Community Subscriptions in Settings. After the routines have been run successfully, the reminder messages are hidden.
Instructions of use, instructions for Thermo HERAcell-CO2-Incubator i i manual en
To remove the support rails, pull the lockspring tab down out of the embossing herracell remove the rail. Note Make sure that power sockets remain freely accessible at all times! Operational Safety Rules CO -incubator. The receptacle for the two device fuses is integral to the socket. Press the auto-start key . Page Suitable instrument: Unless a user action contact with the pressure-sensitive areas and keys occurs within 30 seconds, the system automatically exits the menu and the most recently confirmed setting defaults.
The routine will be disabled after a power reset. Distributor connection for looping through the O gas supply for up to 3 additional devices independent of device type Sensor System O heraclel are integral to the control system of the device. Page 95 The auto-start routine is restarted automatically if the routine is canceled due to one of the following events: Page Chapter 10 Maintenance Note Isothermal container: